[aprssig] House Temperature Probing
scott at opentrac.org
Sat Dec 9 13:50:12 CST 2006
It self-heats only if you're putting a significant load on the regulator.
You can also remote the sensor pretty easily. Or you can load the Dallas
1-wire weather firmware and connect a DS18S20 sensor. I've got one of those
in the garage - it runs about 20 feet from the tracker to my freezer to make
sure the defrost heater hasn't broken again. It's more accurate than the
on-board sensor and doesn't need any calibration. It also isn't affected by
voltage drop on a long cable since it's a digital sensor - it generally
either works or doesn't, with no degradation in accuracy.
If you load the weather firmware with no external sensor connected, it'll
just report the on-board temperature in APRS weather format. That lets you
get graphs on Findu. If you do have an external sensor connected, you'll
still get the on-board temperature in the status text. Check out
wx.findu.com/n1vg-3 for an example.
N1VG-3>APOT01:>Livin' in the fridge! 14.2V 23C
The board itself is at 23 degrees C, and the freezer is at -12 F. The
status text always reports in C, so if you want F from the on-board sensor
you've got to run it in weather mode. On the Tracker2 it's selectable.
As for calibration of the on-board sensor, it has a very linear response so
you only need a single-point calibration. Just check the real temperature,
see what the sensor reports, and enter the difference in the temp adjust
> -----Original Message-----
> From: aprssig-bounces at lists.tapr.org
> [mailto:aprssig-bounces at lists.tapr.org] On Behalf Of Drew Baxter
> Sent: Saturday, December 09, 2006 11:25 AM
> To: aprssig at lists.tapr.org
> Subject: [aprssig] House Temperature Probing
> It's winter again in the Northeast and that means fun things like
> pipes freezing and heaters going out.
> I'd be interested to set up something that reported every hour to
> give me an idea of what the temperature was inside a camp I don't
> often visit throughout the season. If the heat fails, it could make
> a big mess of things.
> I see the Opentracker has an onboard temperature sensor. Does the OT
> get hot in such a way that it could affect the readings from the
> onboard sensor?
> Has anyone tried anything like this? :)
> --Droo, K1XVM
> aprssig mailing list
> aprssig at lists.tapr.org
More information about the aprssig